/ Home / 产品 / Miscellaneous / 配件 Gage Fluids
Replacement fluids for fluid filled manometers. The different fluids are red gage fluid, blue gage fluid, and fluorescein green color concentrate. Dwyer's fluids are furnished in unbreakable plastic dispenser type bottles.
We are using a Durablock inclined manometer with a range of 0-0.25" of water. What is considered a normal response time to steady state? We are seeing times of 1 minute whereas the manual states 'a few seconds'. Gage fluid is well beyond the recommended three years. Do I need fresh (i.e. new) gage fluid? My supply is as old as what's in the manometer.
压力计具有响应时间，这些响应时间取决于管道的长度，因此一分钟的响应时间并不罕见。您可能想要检查您的配件以确保存在目前的CLOG。在固有的属性开始改变之前，花序液有三年的寿命，所以建议更换它。一盎司的液体是p / n a-101。
A 1 oz bottle of A-126 Fluorescein green color concentrate is added to a quart of distilled water. Remove a tubing connector from the manometer and pour the mixture in to mid-point level. Shake the manometer to remove air bubbles and slide the scale so the zero mark is in line with the meniscus of the two fluid columns.
The red gage fluid has a specific gravity of .826. If you dilute it with distilled water you will change the specific gravity and it will not work in instruments that require the red gage fluid.
T: +1 800.872.9141 +1 219.879.8000 F: +1 219.872.9057